Satheesh, K and Jagadeesh, G (2009) Analysis of an internally mountable accelerometer balance system for use with non-isotropic models in shock tunnels. In: Measurement, 42 (6). pp. 856-862.
1fulltext.pdf - Published Version
Restricted to Registered users only
Download (555Kb) | Request a copy
Knowledge of drag force is an important design parameter in aerodynamics. Measurement of aerodynamic forces at hypersonic speed is a challenge and usually ground test facilities like shock tunnels are used to carry out such tests. Accelerometer based force balances are commonly employed for measuring aerodynamic drag around bodies in hypersonic shock tunnels. In this study, we present an analysis of the effect of model material on the performance of an accelerometer balance used for measurement of drag in impulse facilities. From the experimental studies performed on models constructed out of Bakelite HYLEM and Aluminum, it is clear that the rigid body assumption does not hold good during the short testing duration available in shock tunnels. This is notwithstanding the fact that the rubber bush used for supporting the model allows unconstrained motion of the model during the short testing time available in the shock tunnel. The vibrations induced in the model on impact loading in the shock tunnel are damped out in metallic model, resulting in a smooth acceleration signal, while the signal become noisy and non-linear when we use non-isotropic materials like Bakelite HYLEM. This also implies that careful analysis and proper data reduction methodologies are necessary for measuring aerodynamic drag for non-metallic models in shock tunnels. The results from the drag measurements carried out using a 60 degrees half angle blunt cone is given in the present analysis.
|Item Type:||Journal Article|
|Additional Information:||Copyright for this article belongs to Elsevier science.|
|Keywords:||Aerodynamic drag measurement;Impulse facility.|
|Department/Centre:||Division of Mechanical Sciences > Aerospace Engineering (Formerly, Aeronautical Engineering)|
|Date Deposited:||09 Jul 2009 05:21|
|Last Modified:||19 Sep 2010 05:36|
Actions (login required)